Read or Download Agilent 16533A, 16534A Digitizing Oscilloscope (help volume) PDF
Similar nonfiction_6 books
Nanophotonics is a box of technological know-how and expertise in keeping with the manipulation of sunshine with both miniscule buildings, within the similar approach that machine chips are used to direction and turn electric signs. via permitting new excessive bandwidth, excessive pace optoelectronic parts, nanophotonics has the aptitude to revolutionize the fields of telecommunications, computation and sensing.
Within the absence of federal management, states and localities are stepping ahead to deal with severe difficulties like weather swap, city sprawl, and polluted water and air. creating a urban essentially sustainable is a frightening job, yet thankfully, there are dynamic, cutting edge types outdoor U. S.
- Satan's rebellion and fall
- Evaluation of Gas-Turbine Modular Helium Reactor
- The Violet Book of Mentalism
- Distributed Sensor Systems: Practice and Applications
- Schein's Common Sense Emergency Abdominal Surgery, Second Edition
- All-pervading Melodies Sound of Thunder: The Outer Liberation Story of Terton Migyur Dorje By Karma Chagme
Additional info for Agilent 16533A, 16534A Digitizing Oscilloscope (help volume)
The Agilent Technologies 16533A or 16534A can average from 2 to 512 waveform acquisitions but the larger the number of acquisitions, the more time it will take to accumulate all the waveforms you have requested. The following formula is used to calculate the average for each data point: For n between 1 and M. After terminal count is reached (n greater than or equal to M), where: Aven = the average sample value n = the current average number M = setting of # Avgs control (terminal count) Si = the ith sample.
You access it by selecting the Trigger... button at the bottom of the main oscilloscope window. The two selections that are always availabe in the window are Mode and Sweep. Mode specifies the type of condition you want to trigger on. Sweep indicates whether the oscilloscope should wait for the condition (Triggered) or trigger immediately if the condition doesn't show up in 100 milliseconds (Auto). Edge Edge mode sets the oscilloscope to trigger on an edge. You can specify the source, trigger level, slope, and occurrence.
The vertical (V/div) scale control is located under the Channels tab. The horizontal (s/div) scale control is located at the bottom left corner of the oscilloscope window. Scrolling The scrollbar below the display indicates what portion of the current data set you are viewing. Its size shows the percentage of the data you are looking at, and its location indicates the location of the data within the data set. You can scroll through your data set by dragging the scrollbar. You can also use the Delay field, but this may change your acquisition delay as described in “Delayed Triggering” on page 42.